755/808/1064nm Diode Laser Hair Removal Beauty Machine Skin Care Hair Removal u. This machine is Semi conductor Hair removal series, which.
Cooling Technology of Sapphire Epidermis. The laser beam is based on high.
Performance patented light shaping technology, which can be applied to white skin to. Light brown skin with dark black hair. Before operating, Please carefully read and strictly operate the machine according to. Process:808nm laser acting on Skin - Melain of follicle absorbs laser caused heat.Increased sharptly to distroy follicle - Achieving the effect of permanent Hair removal. Hair follicles and hair shafts are rich in melanin, which is distributed between. Sem iconductor laser hair removal applicates specific. Wavelength laser, of 808nm, a the target of melanin, selectively acting on the hair. Follicle, causing its temperature to rise sharply, destroying the hair follicle and its. Surrounding tissues, thereby destroying the hair growth center on which the hair. After absorption of laser light, melanin selectively destroys hairy cells due.
To photothermal effect, fundamentally terminates hair growth and achieves. Permanent hair loss reduction effect, not only avoiding damage to surrounding. Normal tissues, but also shrinkd the pore which always worrying people..
Skin become smooth and flawless. The effective penetration depth of 808nm wavelength can reach the target.
The appropriate pulse duration ensures that the target tissue. Produces enough thermal damage and the surrounding tissue is almost. Unaffected; the proper amount of energy density guarantees sufficient time in a. The energy output is sufficient to damage the target tissue and the. Normal tissue is almost unaffected; suitable epidermal protection measures.
Ensure adequate damage to the target tissue, while the epidermis is almost. Unaffected, thus ensuring the safety of treatment. On the basis of meeting the above design requirement, 808nm lase hair removal. Used the multi-pulse laser by special design to heat hair follicle to 48.The lower energy desity pattern, and keeping a period time(10Hz state) by sliding. Of treatment tool, which makes stem cells lost their growth actvity, thus achieving. Hair of various parts of the body, such as: the hairline of the face, lip hair; beard.
And body hair, chest hair; limb hair, bikini hair, etc. Clean the treatment site, take photos before surgery, shave the hair with long. Hair, leave a little hair root, do not completely scrape it. Select the appropriate wavelength and treatment head according to the.
Treatment item, and select appropriate parameters according to the color depth. Of the customer's lesion tissue and the skin condition of the customer. Select the corresponding skin color according to the customer's skin color on. The operation interface, and then select the energy parameter according to the. Customer's pain and hair thickness, apply a gel at the treatment site, and operate.
4, the clinical end point is recommended: smell the hair burnt, the hair is rolled. Up, part of the hair roots are reddish. 5, after the treatment is completed, postoperative attention. Semiconductor refrigeration + water + air cooling. 10°C to 10°C by cooled sapphire.
220VAC / 50-60Hz(110VAC can be customized). According to the air transportation regulations, the water in the instrument must. Be emptied, and the user should add water after receiving the instrument.
Machine without water, so the damaged bald head should be borne by the user. Mineral water or tap water is prohibited and deionized water (distilled water) is. Regular replacement of cooling water every week.According to the quality of. Check that the handle is good and check the protective sleeve of the handle to. Make sure the sleeve is not bent or pressed. If the casing is seriously compressed.
It will cause water blockage or insufficient water flow, which is not conducive to. The cooling of the instrument, and may cause problems. Before using, first check if the line is connected correctly.
Make sure rated voltage conform to the voltage in your country. Treatment bald head should be placed on bald fixture when not in use.
Hand-held operation should pay attention to protection, so as to avoid bald head. After the machine starts, the water system will be circulated for one minute.Before operation, so that the water system can be completely circulated. Treatment of the bald head is strictly prohibited to directly illuminate eyes. Cold gel must be added during operation. The treatment head must be lightly. Do not leave the skin during treatment to avoid burning the. The strong light hazard prompt refers to the treatment of strong light emitted. From the front of the bald head to illuminate the eyes, which is prone to danger. The operator is required to wear a protective goggle during operation and. Protect the operator from direct exposure to the human eye. The instrument water tank must be filled with water before use.
When different functions need to be replaced or different parameter options. Need to be adjusted, the machine should be in standby state, and the function or. Parameter options can not be replaced in the ready state. [Declaration]: Low temperature area below 0? If the instrument is not used for a long time or needs to change the site, the.Water in the interior (handle and main engine) must be emptied. Try not to place the instrument for more than 3 hours in an environment below. It has been found that water below 0. Will appear to be tumbling up when. And there will be rapid. After the liquid water turns into solid ice, the. Volume increases rapidly, causing certain damage to components such as water.
Pipes, cooling water pipes, and laser waterways. After the customer receives the instrument, it needs to be stationary for 24. Hours in the room temperature of 23-26. Before it can be used. Equipment is not operated as required, and it is damaged due to the customer's.
Own reasons, the company will only provide paid maintenance services. Of the influence of temperature, we hope that customers will pay attention to it.
And follow the requirements for routine maintenance. For white skin (fine/ golden hair) hair removal. For yellow (natural) skin hair removal. For black skin (black hair) hair removal.This item is in the category "Health & Beauty\Shaving & Hair Removal\Electric Shaving & Hair Removal\Laser Hair Removal & IPL". The seller is "motor*pro" and is located in this country: CN. This item can be shipped worldwide.